1. Prior to use read instruction.
2. Do not store in sunlight, high humidity, or low-temperature circumstance.
3. Single-use does not reuse. Treated as infectious medical waste after use.
4. To connect the dialyzer or inner(outer) fistula，please handle under asepsis circumstances.
5. The product is sterilized by Ethylene Oxide with no pyrogen. Do not use if the package has been previously opened or damaged.
6. Focus on the safety of connecting with other medical devices. If blood leaks, nursing staff should remove obstacles and adjust. If fails, replace entire bloodlines.
7. Before use, the Pressure Monitoring Tube must connect with Transducer Protector.